DSC Pan (6.7 x 1.7 mm) and Pan Style Cover Kit
Quantity: 200 Pans & 200 Self-Crimping Covers
Features: Deep pan for higher sample volumes
Uses: Replacement Consumable for TA Instruments Differential Scanning Calorimeter: Q10, Q100, Q1000, Q20, Q200, Q2000, and others.
Proudly Made in the USA!
DSC Large Volume Aluminum Pan (6.7 x 1.7 mm): S-CNAL15
DSC Aluminum Pan Style Cover: S-CNAL16
DSC High Volume Aluminum Pan (6.7 x 2.7 mm): S-CNAL18
Engineering a Better Pan…
This product has been specifically designed and tested to work with TA Instruments brand DSCs. The aluminum impurities were altered/added to stiffen the pan to resist bottom deformation sometimes seen when mechanically crimping high modulus of elasticity (low deformation) samples. Each lot undergoes high speed DSC runs (greater sensitivity) for quality control.
|Temperature Range:||-170 to 600 oC|
|Volume Range:||0 – 40 mL|
|Max Recommended Volume*:||13.33 mL (1/3 of total)|
|Max-Plus Volume**:||20 mL|
|Max Pressure:||1 atm|
*Max Recommended Volume: Samples of this volume or less will generate the best data and are considered safe for system use.
**Max-Plus Volume: Samples with a volume greater than this are highly likely to cause the following errors or damage, each being dependent on the sample composition: pan deformation, inaccurate transition data, spillage/overflow, or cell damage.
Pan and lid sets are crimped to hold the sample in place, but not hermetically sealed.
For an introduction to the Differential Scanning Calorimeter (DSC) and it’s capabilities, watch our free video here.
Engineering Analytics Laboratories (EA Labs) does not manufacture this product at this time. EA Labs resells for a manufacturer, at the set price. EA Labs does not honor discounts or coupons from the manufacturer over the web. To apply a manufacturer’s coupon, please contact EA Labs directly at EngAnLab@gmail.com.
In deference to the Original Manufacturers
“EA Labs and its suppliers are not affiliated with TA Instruments, Waters Corporation, or its affiliates. All claims of compatibility with TA Instrument’s products are the sole claim of EA Labs or its supplies and have not been reviewed, approved, or endorsed by TA Instruments, Waters Corporation, or its affiliates.”